General Safety Precautions To prevent possible electrical shock, fire, personal injury or the device damage: • Carefully read user’s manual. • Do not place sensor on the objects which exposed to high voltages. These voltages could cause personal injury or death. • The Analyzer could not be used in potentially explosive environments. • Take measures to prevent cables and straps become entangled by rotating part of machines at measurement site. • Do not expose 107VF parts to heavy impacts, high humidity and extreme temperature. • Do not try to open the display unit – this can damage the system, and your after-sales service warranty will come void
Overview The 107VF Vibration Analyzer (Device, Analyzer) is a compact yet powerful, vibration analyzer designed to measure overall vibration parameters, FFT spectrum analysis of the rotating machinery, immediate evaluation against ISO 10816 standard, condition monitoring by route based measurements and data collection. Route files and data files exchange via email makes it ideal for data collection at remote sites. Simple in use, with free firmware upgrades, comes with data management and reporting software.
Kit Content The 107VF kit includes: ▪ ▪
▪ ▪ ▪ ▪
107VF display unit; AC102-1A accelerometer, incl. cable 1.8m, magnet for curved surface mount; USB wall charger; USB cable; CD with ConSpect software and User’s Manual; Carry case.
Specifications Inputs – IEPE or charge type accelerometers with known sensitivity, switchable. Optical RPM transducer with IR pyrometer sensor (optional) AD conversion – 24 bits Frequency range – 1…10000 Hz
Temperature measurement range –
-70°C to 380°C
FFT spectrum resolution – 400, 800, 1600 lines Data storage – 4GB micro SD card PC interface – USB Display – color, sunlight readable 128x160 dots Battery – Li-Po rechargeable, up to 8 hrs continuous operation Dimensions – 132 x 70 x 33 mm Weight – 150 g
Measurement functions Vibration mode – analyzer measures overall level of vibration acceleration, velocity and displacement and FFT spectrum, route or off-route measurements. Tachometer – analyzer measures speed of rotation by means of contactless optical sensor. The measurement result is displayed in RPM and Hz. IR thermometer – contactless measurement of object temperature. The measurement result is displayed in °C and °F.
Operation Keyboard – press and hold for 3 sec to turn device ON, short press to turn OFF – Enter, confirm selection, start measurement – navigation arrow keys – Menu – backspace, quit – option key
Settings This menu is used to: - set Date/Time - Sensors setup - Units setup - Auto OFF delay setup
Use arrow keys
to set date.
Hold then press decrement/increment. Confirm by
when correct date is set.
keys to set minutes and hours.
Use key to switch focused field. Focused field is indicated by red frame. Confirm by
when correct time is set.
Use keys to choose sensor, which will be used for measurements. Drop down menu offers two types – IEPE or charge type sensors to choose from. Confirm choice by
Type, S.N. and Sensitivity fields are editable. Use
key to choose field to edit.
Then use arrow keys
to edit the field value.
Units This option is not implemented yet..
Auto OFF Use Press
keys to set auto OFF delay (minutes). or
key to confirm and quit menu.
Vibration Analyzer measures vibration Acceleration, Velocity and Displacement. In ISO 10816 mode measurement result is compared to the built-in table of vibration severity grades according to ISO 10816-3.
keys to choose measurement mode.
Vibration measurement settings
key to enter Settings menu.
to choose parameter to setup.
to change parameter value.
Low Freq – lower frequency limit. Can be set to 1, 2, 10 Hz. Hi Freq – upper frequency limit. Can be set: - from 200 to 10000 Hz for Acceleration; - from 200 to 5000 Hz for Velocity; - from 200 to 800 Hz for Displacement; FFT lines – FFT spectrum resolution. Can be set to 400, 800, 1600 lines. Trigger – not implemented yet.. Averaging – measurement averaging. Can be set in range of 0 to 64. Zero means that averaging is OFF. Window – weighting function. Can be set to Hanning or Rectangular.
Taking measurements Choose vibration parameter e.g. Velocity, edit settings if needed, then press measurement.
key to start
When measurement is running: Use key to toggle FFT spectrum / waveform display. Press key to stop/resume measurement.
When measurement is stopped: Press
key for Options:
Save.. – to save measurement data. Press to proceed.
Format – Linear/Logarithmic amplitude display. Use
to change parameter value.
Zoom – frequency axis display zoom change. Use parameter value
To save measurements Press Press
key to stop measurement key for Options
Choose Save.. and press
Device will enter My documents menu Browse to the destination folder, then press measurement.
Device writes two files at a time – FFT spectrum file and waveform file. Device remembers path to the last written files. To create new folder – press default name for new folder.
key. Date/time stamp is used as a
To create folders with meaningful names – connect device to the PC via USB as external flash drive, then create folders using PC keyboard.
Route based measurements • •
Using ConSpect software create route file and download it to the device Go to Documents menu, move cursor to the route file and press
Use to browse route points
Attach sensor at the measurement point and press key. Device takes measurement with preset parameters and saves files to proper destination folder
Tachometer (107VF-T, 107VF-T2 only) Connect optical probe to the device Enter Tachometer menu Aim optical probe to the rotating machine part with attached reflective tape. Press
key to start/stop measurement.
Device displays measurement result in RPM and Hz
Thermometer (107VF-T2 only) Connect optical probe to the device Enter Thermometer menu Aim optical probe to the machine. Press